Editor’s Note: When my Geiger counter kit refused to count (Click Hereto see details), my friend David Ashton in Australia kindly offered to take a look at it for me. He also kindly offered to document what he did in the hopes that it will help others with the same kit. So now let’s hand over to David…
When I received the counter I noticed it had a wire broken on the speaker. Max advised me that this had been connected when he assembled it, so I assumed it had broken off due to jarring in transit. I did not immediately reconnect it, which in retrospect was a bad idea! I first set about probing around with my digital multimeter (DMM) and oscilloscope. First let’s look at the schematic of the counter as illustrated in Figure 1.
Let’s start with IC1, which is a 555 Timer. This chip is intended for use in a variety of timer, pulse generation and oscillator applications. As an aside, the design for the 555 was proposed in 1970 and the first production chips were released in 1971, so this chip design has been gamely working away for 40 years now, which is rather impressive when you come to think about it.
Anyway, IC1 is a 555 in astable mode. It feeds Q1 which drives transformer T1. The secondary voltage of T1 is rectified and via R3 supplies the Geiger tube GM1. This tube will have a very high impedance in the absence of any radiation, but when any beta or gamma radiation is present this will ionize the low pressure gas inside it and the tube will “strike” much like a neon bulb. This will result in a short negative-going pulse at the + terminal of GM1. This pulse is used to perform two actions:
Switch on Q2 to produce a click in the speaker SPK
So far so good, but on close examination I realized that there are a few strange things about this circuit as follows:
The 555, though in astable mode, does not employ the standard astable configuration, which uses a resistor Ra from the supply to pin 7, another resistor Rb from Pin 7 to pins 6 and 2, and then a capacitor to ground. In this circuit, Rb is effectively zero. Rb governs the Off time (negative-going) pulse time, so with no Rb this time will be of extremely short duration.
The output of T1 is rectified by diode D1 but there is no smoothing capacitor at its output, so the GM tube will receive a pulsed DC.
Although Q3 and Q2 are referenced to the negative side of the battery, the high voltage part of the circuit is not DC-referenced to the low voltage part of the circuit. There will be some capacitive coupling from the T1 secondary to T1 primary, however, and the LED drive circuit is capacitively coupled to the GM tube anode by a 10 pF capacitor.
There is no limiting resistor in series with the LED as is common practice. Possibly the LED is a current limited one and does not need one, but another correspondent with a slightly different model counter had a place for one on his PC board.
Max’s initial complaints were as follows:
The speaker did not click at all, even when the circuit was brought close to a radioactive source.
The LED stayed on all the time.
Max had confirmed that he got a “click” from the speaker when he shorted the tube (simulating a radiation “event”). My initial theory was that there was not a high enough voltage to get the tube to ionize properly, so my first test was to check the voltage across the tube.
The result was around 79 volts, which is far too low. I had noticed the unusual configuration of the 555 oscillator so I tried putting in an Rb resistor to make the duty cycle of the pulses a bit longer. This entailed cutting some tracks on the PCB around the 555, but when I did it the voltage on the tube looked even lower. Looking at T1 output with a scope showed a very short duration “ringing” pulse output. The circuit description said that T1 provided a high voltage output due to its turns ratio. I am inclined to think that there is more of a “flyback” effect here, as the pulses are extremely short and lengthening them seems to give less voltage at the output.
Figure 2. Scope trace of T1 Output
Hor: 2mS / div, Vert: 100V / div (10v + x10 probe)
This was taken after I had put a 1nF smoothing cap on the other side of D1, which accounts for the difference between the positive and negative values.
The 555 operates at about 300 Hz. At one point I connected my DVM while my scope was already connected at T1 output and the amplitude decreased severely. I came to the conclusion that because of the high impedances of the tube and the very low duty cycle of the oscillator, even a 1 Megohm load from a DVM was probably enough to overload the HT supply to the tube and drag it down.
One of my theories all along was that the HT supply should be smoothed, so I tried various capacitors at the Cathode of D1. All seemed to improve matters – the HT went up to a bit over 100 volts on my DVM – but by this time I was convinced that my DVM was too much of a load for the HT circuit. About this time the one remaining wire on the speaker gave up the ghost and the speaker fell off.
Editor’s Note:The speaker fell off? The SPEAKER FELL OFF? My lower lip is quivering and a little tear is rolling down my cheek … my poor little baby!
I attached it to the board with double-sided tape and re-soldered it to the board with new wires. Once I had done this, and without any test equipment or smoothing capacitor in the circuit, I switched the counter on.
It made a click when I switched it on… then a few seconds later I got another click… then another… then some more at random intervals. It seemed to be working! The LED was still on all of the time, but it did flash slightly brighter when a click came. If I connected my DVM or scope onto the HT circuit, the clicks stopped. This confirmed my suspicion that even a 1 Megohm DVM or Scope input loads the HT line too much. Coming from the valve (vacuum tube) era, I should have known better!
The circuit was now operating without any modifications – I had undone my changes to the 555 circuit and did not have a smoothing capacitor on the HT. So why had it not worked for Max? He was getting clicks when he shorted the tube, so the speaker must have been connected. Some correspondents have reported high sensitivity to humidity, and possibly (being in Alabama in storm season) this may have been a factor. More than that I cannot say, I found no obvious cause that would have stopped it working.
I still had the problem with the LED being on all the time. One of my theories for this was that the unsmoothed HT pulses were getting through to the LED driver circuit and switching it on. So a smoothing capacitor should improve this. Other circuits have used a variety of capacitors – 2.2 nF being a common value. I had a 1 nF and 4.7 nF. Both improved the LED condition – it got a bit dimmer – but it did not go off. There did not seem to be much difference between to two caps, so for size reasons I stuck with the 1nF. The fact that the circuit works with no smoothing cap shows the high impedance nature of the GM tube – its own capacitance and stray capacitances in the circuit do smooth the HT to quite a large extent. The following trace is with a 1nF capacitor.
Figure 3. Cathode of D3.
Hor: 2mS / div Vert: 100 v/div (10v / x10 probe).
Note that even this loads down the HT considerably and these are not the true voltages or waveforms in the circuit.
My other theory was that possibly – since the LED had no current limiting resistor – even short pulses of high current in the LED were causing it to be much brighter than it would be if the current was limited. So I put a 470 ohm resistor in series with the LED. There was a very slight – if any – improvement, but obviously this was not the problem.
I also tried increasing the value of R3 which supplies the HT to the GM tube. Most circuits I have seen use anything between 2 and 10 Megohms for this, rather than the 100K used here. I changed R3 to 4.7 Megohms. It did not change the LED operation much, if anything, but it did improve the intensity and quality of the “click” from the speaker.
The LED driver transistor Q3 is connected to the GM tube anode via a 10 pF capacitor and a 10 Megohm resistor. So any positive-going pulse of more than a few volts would turn the transistor – and the LED – on to some extent. I experimented with a few megohm-value resistors across the Base-Emitter of Q3. There was an improvement but not a complete turn off of the LED. I also tried – separately – a couple of capacitors – a 10 pF and a 39 pF I think – again with not much improvement.
To have a look at the waveforms I was dealing with, I put my scope across the BE of Q3. There was an immediate and dramatic improvement. The LED stayed off and only flashed when the speaker clicked. It was then a simple matter to measure my scope probe – it was 1 Megohm and about 72 pF. I tried a 1 Meg resistor and a 68 pF capacitor across the BE of Q3 and it worked as well as the scope probe. I managed to capture an event along with the power supply pulses and there was a significant difference in both the amplitude and duration of the power and event pulses:
Figure 4. B-E of Q3. Power supply plus event pulses.
Hor: 2mS / div Vert: 1v / div.
This is with 1 Megohm + 68pF already across the BE of Q3 so the scope effectively makes this 500K + 140 pF approx, which will probably attenuate both pulses lower than actual values. The event pulse does have a strong negative component as well, which is not visible in this trace.
My one large problem came in testing the circuit. I tried all sorts of things that might have some level of radioactivity. I had an old glow-in-the-dark compass but that did nothing. I tried my smoke detector – but these emit primarily alpha particles which have very little penetrating power (they have difficulty getting through a sheet of paper, let alone the glass of a GM tube).
Eventually I took it to a Nuclear Medicine practice in a neighboring town (they do radiotherapy and various diagnostic tests using radioactive substances) and they very kindly let me have access to a radioactive source. Below you can see a short movie I took of this. In my workshop at home I was getting 10-20 counts (clicks / flashes) per minute which I gather is pretty typical for background radiation. When I first switched the counter on in the lab of the nuclear medicine place, I was getting 2-3 times that. As I approached the radioactive source (which I was told was an 8 GigaBequerel Technetium source giving primarily gamma radiation) the clicking and flashes rapidly increased. At the nearest point, the clicks had merged into a buzz, and the LED almost extinguished. I presume this was due to the short time between event pulses causing them to reduce in amplitude as the tube did not have time to recover fully (“quench” is the technical term) in between pulses. At any rate. you’d certainly be aware of it long before the radiation got to this level!
One correspondent suggested that the “No-salt Salt” that you can buy (which replaces a lot of the Sodium Chloride with Potassium Chloride) is mildly radioactive. I tried this just before I sent the counter back to Max and it does indeed work. What I bought was “Lite” salt which is about 50% NaCl and about 50% KCl.
Figure 5. Geiger counter under test with potassium chloride salt.
I took three 1-minute readings with just background radiation followed by three with the (cardboard) Lite salt container right against the tube. The results are as shown in Table 1. As we see, there’s not a huge difference, but there is definitely an increase from the background and does provide a cheap and easy (and tasty) test.
Table 1. Geiger counter under test with potassium chloride salt.
Click Herefor quite a good summary of common radioactive materials.
Humidity: Some users reported high sensitivity to high humidity. This was difficult to test as my workshop was usually around 10 degrees C, so breathing on the counter probably resulted in condensation on the components and did stop it functioning. However, the problem usually cleared in a couple of minutes; faster if hot air was blown on it. Non-condensing humidity should not be a problem.
Note: This counter will detect Beta Particles (electrons) and Gamma rays. Alpha particles (helium nuclei) will not pass through the glass and metal walls of this tube and will not be detected. Smoke detectors emit mainly alpha radiation and mine did not register on this counter at all. Some readers say that they have been able to detect smoke detector radiation, though.
I have documented the modifications I made below in case anyone wants to try these on their units. Chaney make some similar kits and if yours is not the same as Max’s (C6979) you should be able to work out the changes for your kit using the schematics below.
Below is a copy of the schematic with my modifications shown in RED:
Figure 6. Modifications to Geiger counter Schematic as described in text.
Existing and modified component values are shown below.
C4 is soldered to D1 cathode and the Jumper wire as shown. C5 and R7 are mounted in new holes on the board as shown in the PCB diagram below. R7 may be bent over R6 as shown to avoid it sticking up above other components. R3 is changed from 100K to 4M7. The current limiting resistor for the LED, R8, is mounted on the track side of the board as shown below.
Figure 8. Modifications to PC Board.
Two pairs of holes are drilled adjacent to the copper tracks coming from the Base and Emitter of Q3 as shown, and C5 and R7 are mounted in these holes (C5 adjacent to Q3). Careful siting of the holes enables the component leads to be soldered to the tracks, or the leads may be bent over onto the tracks.
The track from the collector of Q3 to the LED is broken as shown and a 470 ohm resistor is soldered to the 2 pads and mounted horizontally on the board as shown. This modification does not directly eliminate any problems and may be omitted if desired and/or if operation is satisfactory without it.
Testing with varying battery voltage
Voltages higher than the nominal 9V battery voltage (up to around 11V) showed no change in operation. Lower voltages (down to around 7V) also showed no appreciable changes. Voltages lower than 7V resulted in some power supply buzz in the speaker, though the counter did continue to work.
My aim was just to get this Geiger counter working, though the temptation to make major changes was great! Having read up a bit on Geiger-Muller tube operation, most tubes need a voltage of around 400V for optimum operation, with a “Plateau” about 40V either side of this where operation will be satisfactory and repeatable.
Although it is not really possible to measure the voltage at which this particular unit operates, it is fairly obvious that the voltage on the tube will vary with varying battery voltage. Maxim produces an IC that can be used to generate a suitable regulated supply (Click Here for details) and were I to build a counter from scratch I would use something like this. If I was doing a lot of this, I’d be tempted to get or develop and make some test equipment that could measure high voltages at very small loads. With the very high input impedance of Intersil’s ubiquitous ICL 71x6 chips this shouldn’t be too difficult.
Most correspondents who had the same or similar kits had problems with the LED staying on, and this is the main area in which the folks who designed this kit appear not to have done their homework. The fix is fairly simple and wouldn’t increase the cost of the kit much (obviously for this kit that is a prime consideration). That said, once the kit is working properly its performance is pretty good – it is fairly sensitive and works satisfactorily over varying battery voltages.
Many readers commented on Chaney’s poor support and refusal to acknowledge any problems. With what I have learned from this, I’d rather build my own I think. Or maybe I should get into the kit business!
I must apologies to Max and his many correspondents for the length of time taken to get this information – and the kit – back to him. I had 10 days off around Easter in April and hoped to have the counter to get stuck into during that time. Thereafter I was very busy at work with some projects and could not devote the time to it that I would have wished. It arrived a few days after I was back at work, of course. Max’s shipping really was by ship I think!
My thanks – and admiration – must go to all the correspondents to Max’s pages on the subject, There were some very detailed analyses of the kit – far better than I have achieved here. I tried not to be initially influenced by these posts, though most of them seem to have come up with very similar – possibly better – solutions than me. Thanks all for the inspiring input.
Here are some links to useful information and other posts:
Click Here for a summary of tube characteristics for several Russian tubes (although this appears at first glance to be in Russian, English translations are provided also).
EETimes reader pchow documented his mods – very similar to mine – very nicely (Click Here). Nice work, Peter.
EETimes reader finevlad gave links to other schematics (Click Here and Here). These were very useful in seeing how totally unrelated circuits drove the tube.
EETimes reader spyderjacks documented his excellent debugging (Click Here), including great scope prints – I wish I had a scope like that! I’m fairly ashamed of my antique….but it works…
Thanks again to EETimes reader gwasic for the suggestion about the No-Salt Salt.
And to all the readers who took the trouble to comment on Max’s original blog… my apologies once more for keeping you in suspense for so long.
Found this thread just this month. Bought and assembled a very similar circuit from Electronic Goldmine. While the circuit does work, I'm having trouble understanding how some parts do indeed work. Specifically the Q3 circuit. Q3 takes a positive pulse to turn on, but a gieger tube pulse will appear as a negative pulse, at least the leading edge of it. So, it would seem that Q3 only turns on when the tube recovers (trailing edge) and/or the high voltage line charges back up. I suspect the recovery time may be faster than the charge-up time. Has anyone investigated this? Hope someone besides me is still interested in this.
This has been an interesting read for me. As someone who has also created a Geiger kit, I can understand some of the oddities of trouble shooting a HV circuit.
The HV will travel very well through a poor conductor - excess flux on board, etc. I even discovered (unintentionally) that black foam board is conductive enough to short out the HV!
Also, since the current available is so small, measuring it with a DMM will load it down so much you won't get a good reading, and the HV might stop working while the DMM is connected.
At the risk of plugging a competitive product, I have written instructions that apply to measuring HV in Geiger circuits here . . .
The should apply to any HV circuit.
Thank you so much for making this available. I got a new kit in today, revision 04/12/11LW, which did something similar. The schemos include a 100R resistor for the LED but everything else is the same. The sensitivity was almost nothing, probably a click every 3 minutes or so. I noticed that if I touched the + end of the tube with a single voltmeter probe, it stabilized and I got actual background level, about 25CPM. To my surprise I had disk shaped 1000pF/500V caps in a drawer (model military suitcase) and fitted one of them standing vertically on the outside of the transformer with one pin between T1 and Q1 to the jumper over to the (middle) case (-) of the GM tube, the other pin along the edge of the board to the cathode of the diode. I also changed R3 to 1MOhm (biggest I had in throughhole). The board now works beautifully with no double clicks.
(Repost from the blog)
I must have been thinking of another tranny I was using recently. And you'd never get this info off the kit maker. I've seen kits - and boards - where the maker has sanded the markings off the ICs to stop people copying their design...
As a bit of an aside, I just bought a few LCD voltmeter modules cheap from my local Dick Smith (Aussie equivalent of Tandy). They say they have 100 Megohms input impedance. So it should be possible to make a 2KV meter out of one of these with an input R of about 1 gig-ohm. Anyone know where I can get some 100 meg resistors??
Hmmm....smart maybe (even that is stretching it) but genius, no. I regularly surprise even myself with my own stupidity....
There is no information on T1 in the kit documentation and none of my photos show any markings. As I recall there were some markings on the transformer (on the tape covering the windings) but not sure what they were. Max??
D1 is indeed a 1N4007 - this has 1000v max reverse voltage so quite adequate for this job.
If you're thinking of building one of these yourself, if you can get an inverter transformer from a small fluorescent lamp it should do. A turns ratio of about 1: 50 would be needed (9V to 400V) but do have a look at the Maxim app note at
which is for a regulated supply. Needs a few more (cheap) components, but no transformer (it's a boost switchmode supply with a voltage multiplier).
Thanks Jay. The lack of DC coupling between the HV/LV did worry me as well. And I was tempted to try connecting them together. However you will note that the speaker stage is DC-coupled to the tube (ie uses the capacitance between the transformer windings for AC coupling) and if I'd DC coupled it then that would quite possibly have had other problems.
I DID try putting a large-ish cap (1 uF) between the HV/LV grounds and it did not make much difference.
The diode across BE of Q3 is a good idea. I'm not sure if the led going off at high counts was due to the charge on C3 or because of the capacitance (68pF) I put across the BE. As you can see form the trace above, the "event" pulse and the power supply pulses are quite different at low count rates, but at high count rates I think the event pulses would merge a bit and reduce in amplitude, so my cap would attenuate them to the point where they no longer switch Q3 on. So I'd also be keen to try Pchow's solution (lower R and no C) with a high rate source.
As I said, were I designing form scratch I'd do it differently, but in this case I wanted to get the existing design going with the minimum of modification. I had such fun with this that I might buy myself a tube and try that.....
HURRAY!!! My (now-working) Geiger counter has just arrived back from David.
It’s on my desk clicking away randomly – my (mildly) uranium-enriched glass marbles do seem to increase the count a bit – but I need to time things like David did to get a good feel for it.
I did just borrow a 5 uSv radioactive sample from the folks in the next bay and the little scamp started clicking and flashing furiously.
I also want to try it with that "No-Salt Salt" David used.
I’m heading out to DAC this weekend – so I’ll take the video and write the follow-up blog when I get back.
THANK YOU DAVID!!!