Breaking News
Page 1 / 2   >   >>
User Rank
Blogger
Re: interesting
11/5/2013 9:33:55 PM
NO RATINGS
A voltage is developed along the temperature gradient in a wire. That could be the entire length, but not necessarily.

Say you have 1m of wire. The hot end is at temperature T. The other end at room temperature. Suppose that at 10cm from the hot end, the temperature is also at room temperature such that from 10 cm to 100 cm of the wire is at room temperature. The voltage will appear only from the hot end down the wire for 10cm. Assume that there is no voltage drop for the remaining 90cm, the the voltage from the hot point to the 10cm point will be the same as from the hot point to any point along that 90cm.

User Rank
Blogger
Re: interesting
11/5/2013 9:32:58 PM
NO RATINGS
A common installation in temperature controllers is to place a 1MΩ resistor from one thermocouple wire to a positive voltage rail. That way if the circuit breaks, the indication goes to a high temperature, shutting down any heaters. That prevents a runaway condition.

User Rank
Blogger
A similar measurement article
11/4/2013 8:47:03 AM
NO RATINGS
See Aubrey Kagan's article Current Measurement, Part 1. Four-wire conecpts there are similar to measuring temperature with an RTD.

User Rank
Blogger
Re: simple principle
11/4/2013 8:39:07 AM
NO RATINGS
tomii, presumable, if you wre to build this circuit, you've calibrate it using some other instrument that is NIST traceable. At the very least, you'd compare the measurement circuit against something you trust, NIST traceable or not.

Calibration is all based on trusting the device you're using as a standard. I've been to stadarnds labs where they people compare the latest temperature-measuring instruments, with 4-wire RTDs and 20+ bits of resolution against glass thermometers. Sounds crazy, right? But, the glass thermometers was the trusted instrument. The high-end digital device was an unknown.

User Rank
Blogger
11/4/2013 8:34:17 AM
NO RATINGS
@David, Thanks for pointing out the link to page 2 of the article. I've updated the link at now it goes to page 1.

User Rank
Blogger
10/29/2013 11:41:13 PM
NO RATINGS
User Rank
Blogger
Re: interesting
10/29/2013 11:19:20 PM
NO RATINGS
Hi Caleb.  I don't have an engineering degree either - so I am glad I am in such good company!!   but I digress....   Temperature measurement.....  I have been doing some articles for Max recently on the PICAXE MCUs.  The first one is posted, the second will be up soon but in the third, which I am working on at present, I am dealing with the Dallas-Maxim DS18B20 digital temperature sensors.  Accuracy of 0.5 deg C, from -55 to +125, and the interface to the PICAXE, and reading them,  is REALLY easy,  Not in the same league as the thermocouple system in the EDN article, but for a quick easy temp measurement I don't think you could get easier.  I'll try and ping you again when the relevant article is up.

User Rank
Blogger
Re: interesting
10/29/2013 10:32:07 PM
NO RATINGS
@Caleb, ping me sometime about temperature measurement. I came from that business.

We have a nice technical discussion going on at EDN. My article there gets into more detail.

User Rank
Blogger
Re: simple principle
10/29/2013 1:44:40 PM
NO RATINGS
I am iinclied to believe their accuracy is likely a bit lower than that, or that they have used their calibration source to eke out some extra accuracy.

There are a lot of sources for error in that circuit, many of which are hard to calibrate out (without NIST-traceable hardware), and that sort of defeats the point of it being "low cost," I think.

No to rip on them, I think it's a neat little circuit/idea, but repeatability but it's good to know "going in" what you hurdles you might expect...

User Rank
Blogger
Re: interesting
10/29/2013 1:40:21 PM
NO RATINGS
I'll say that the posted article is quite a bit shorter than the original ..  Part of that was handled by the link to the older thermocouple article.

I think every one of us has gotten it "wrong" before (if not still).  Extra sources of error also come from the calibration curve of that thermistor (the PT1000) and the junction of diddimilar metals at contact points (e.g. thermocouple lead to copper PCB trace).  Dissimilar metals in contact with each other present a "galvanic" potential (this is how a battery works, right?), where other impurities/humidity act as the electrolyte.  (thus the source of "galvanic corrosion")

Also, the variance in "on channel resistance" over temperature of the analog mux may well be a significant percentage of error in this circuit (from the part's datasheets)

But I ramble...

You're welcome ;)

Page 1 / 2   >   >>

Flash Poll

#### Datasheets.com Parts Search

##### 185 million searchable parts (please enter a part number or hit search to begin)
Frankenstein's Fix, Teardowns, Sideshows, Design Contests, Reader Content & More
Betajet">

The Circle – The Future's Imperfect in the Present Tense 1 Comment
The Circle, a satirical, dystopian novel published in 2013 by San Francisco-based writer Dave Eggers, is about a large, very powerful technology company that combines aspects of Google, ...

latest comment Physiker

I'm not sure if I read more than most folks or not, but I do I know that I spend quite a lot of time reading. I hate to be idle, so I always have a book or two somewhere about my person -- ...

latest comment TonyTib

No 2014 Punkin Chunkin, What Will You Do? 2 comments
American Thanksgiving is next week, and while some people watch (American) football all day, the real competition on TV has become Punkin Chunkin. But there will be no Punkin Chunkin on TV ...

latest comment MeasurementBlues

As every developer knows, there are the paper specifications for a product design, and then there are the real requirements. The paper specs are dry, bland, and rigidly numeric, making ...

latest comment MWagner_MA
Special Video Section
The LT8640 is a 42V, 5A synchronous step-down regulator ...
5:52
The LTC2000 high-speed DAC has low noise and excellent ...
5:14
How do you protect the load and ensure output continues to ...
6:54
General-purpose DACs have applications in instrumentation, ...
7:34
Linear Technology demonstrates its latest measurement ...
10:29
Demos from Maxim Integrated at Electronica 2014 show ...
4:17
Bosch CEO Stefan Finkbeiner shows off latest combo and ...
5:59
STMicroelectronics demoed this simple gesture control ...
2:15
Keysight shows you what signals lurk in real-time at 510MHz ...
4:17
TE Connectivity's clear-plastic, full-size model car shows ...
3:42
Why culture makes Linear Tech a winner.
6:08
Recently formed Architects of Modern Power consortium ...
6:39
Specially modified Corvette C7 Stingray responds to ex Indy ...
8:52
Avago’s ACPL-K30T is the first solid-state driver qualified ...
2:42
NXP launches its line of multi-gate, multifunction, ...
4:56
Doug Bailey, VP of marketing at Power Integrations, gives a ...
9:28
See how to ease software bring-up with DesignWare IP ...
5:19
DesignWare IP Prototyping Kits enable fast software ...
4:01
This video explores the LT3086, a new member of our LDO+ ...
7:03
In today’s modern electronic systems, the need for power ...
5:48