Another key issue is peak current. Almost all wireless-based sensor networks rely on some level of duty-cycling to save power and restrict the usage of radio space, which generates peaks in the current consumption profile of the sensor. Low peak current consumption in the radio transceiver reduces constraints on the wireless sensor’s power supply. Output impedance is also important, as it has a major effect on power amplifier (PA) power consumption. Most radios have output impedance below 100 Ohms. Low impedance is only required for high-output-power, long-range applications, however, and results in up to five times higher current consumption than higher-output impedance options that are more suited for short-reach wireless interconnect applications. Overall, assuming a similar receiver sensitivity and PA efficiency, a high impedance 900 MHz radio would use only 1 mW in its PA to achieve the same range as a 50-Ohm 2.4 GHz radio using 25 mW to 40 mW of power.
The choice of carrier frequency also influences power consumption. The two available options within the medical (ISM) radio band are 2.4 GHz or sub-GHz frequencies. The most prevalent 2.4 GHz protocols are Wi-Fi, Bluetooth and ZigBee. In low-power and lower-data-rate wireless medical monitoring applications, however, sub-GHz wireless systems offer several advantages, including reduced power consumption, as well as longer range for given power.
The Friis Equation quantifies the superior propagation characteristics of a sub-GHz radio, showing that path loss at 2.4 GHz is 8.5 dB higher than at 900 MHz. This translates into a 2.67 times longer range for a 900 MHz radio since range approximately doubles with every 6 dB increase in power. To match the range of a 900 MHz radio, a 2.4 GHz solution would need greater than 8.5 dB additional power. Another benefit of sub-GHz carrier frequencies is that they reduce the risk of interference from airways that are crowded with colliding 2.4 GHz Wi-Fi, Bluetooth and ZigBee signals used in in everything from wireless hubs and computers to cellphones and microwave ovens. Sub-GHz ISM bands are mostly used for proprietary low-duty-cycle links and are not as likely to interfere with each other. The quieter spectrum means easier transmissions and fewer retries, which is more efficient and saves battery power.
Furthermore, the narrower sub-GHz bandwidth creates higher receiver sensitivity and allows efficient operation at lower transmission rates. For example, at 300 MHz, if the transmitter and receiver crystal errors (XTAL inaccuracies) are both 10 ppm (parts per million), the error is 3 kHz for each. For the application to efficiently transmit and receive, the minimum channel bandwidth is two times the error rate, or 6 kHz, which is ideal for narrowband applications. The same scenario at 2.4 GHz requires a minimum channel bandwidth of 48 kHz, which wastes bandwidth for narrowband applications and requires substantially more operating power.
Carrier frequency also has a major impact on the average power budget at the network level. Zigbee and Bluetooth offer highly sophisticated link and network layers, but these stacks account for up to 50 to 75 percent of the radio power consumption, with larger overheads. For ultra-low-power systems, the “one size fits all” standardized option is rarely the optimum solution. Instead, designers developing solutions for ultra-low-power applications should consider using the protocol best suited for their need. Finally, link data rate is one of the most important factors influencing power consumption in duty-cycled wireless links. The average power is almost inversely proportional to the link data rate; for instance, a 100 kbps radio will consume almost half the power of a 50 kbps radio for the same payload.
When comparing RF transceivers, “energy per bit” is a better indicator of power efficiency than current consumption. But high data rate radios are often those with the higher peak currents, and these are highly undesirable for most small batteries as they result in large, leaky, storage capacitors. Each of the aforementioned factors is critical for applications where power is at a premium and payload is greater than 10 bits/s. Whereas previous body-worn wireless sensors could only be used for slowly varying parameters, new RF technologies can be used to help observe more rapidly changing physiological parameters, such as heart and brain electrical activity or blood oxygenation, that require data rates on the order of 0.5 to 5 kbit/s to extract meaningful waveforms. One example of a solution that delivers this level of performance is the ZL70250 transceiver from Microsemi – see figure 2. Housed in a chip-scale package measuring about 2- x 3-mm, it has standard 2-wire and SPI interfaces for control and data transfer using any standard microcontroller. The microcontroller’s analog-to-digital converter (ADC) connects to the ultra-low-power analog front-end device. Combined with the ZL70250 transceiver, the resulting solution can be used to develop a wireless ECG solution that can run continuously from a CR series coin cell for up to a week. Similar power efficiency can be achieved with such devices as a 3-axis accelerometer or pulse-oximeter for patient respiration measurement, as well as a variety of other wearable health monitoring platforms.
Figure 2: Block diagram of a typical wireless sensor based on the ZL70250. Click image to enlarge
The advent of micro-power batteries along with advances in ultra-low-power transceiver technology are making it possible to build smart, flexible and smart wireless sensors. Proper transceiver selection is critical for addressing a variety of key design issues so that wearable wireless medical devices can perform continuous monitoring of bio-signals for long periods using a single, low-cost battery.
About the author:
Reghu Rajan is Technical Marketing Manager in the Wireless Machine to Machine group at Microsemi.
Courtesy of EETimes Europe ------------------------
If you found this article to be of interest, visit Medical Designline
where you will find the latest and greatest design, technology,
product, and news articles with regard to all aspects of clean
technologies. And, to register to our weekly newsletter, click here.